For better program readability, you can send the long form. Page Index OUP Warranty options, No part of this document may be extended support contacts, product photocopied, reproduced, or translated to Page Index verification equipment As shown in the following figure, the power supply has three groups of status registers; Operation, Questionable, and Standard Event. With analog programming enabled, you cannot program the output voltage or NOTE current using the front panel knobs or the remote interface. Insert the base through the outside opening of the AC input cover.
|Date Added:||17 July 2004|
|File Size:||55.21 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Insert the base through the outside opening of the AC input cover. Warranty options, No part of this document may be extended support contacts, product photocopied, reproduced, or translated to Connect the Agilent A across the shunt.
Keysight Technologies / Agilent Technologies N5700
The data may be numerical or character string. With analog programming enabled, you cannot program the output voltage or NOTE current using the front panel knobs or the remote interface. With the output off, adjustments can be made to the power supply or the load without shutting off J5700 power. Appendix C Service Error Messages As further explained in this section, the following factors should be considered when selecting wiring to connect the load to the power supply: Output Commands 5 Language Agilsnt Output Commands Output commands enable the output, power-on, and protection functions.
AGILENT TECHNOLOGIES N SERIES USER MANUAL Pdf Download.
Returns the value of the operation event register: To print the manual completely, please, download it. Page AC peaks of the output current ripple, several DC measurements should be made and averaged. Error List The following table documents the various error messages that the power supply supports: In constant current mode, the power supply regulates the output current at the selected value, while the voltage varies as required by the load.
Page of Go.
Programming Examples 6 ‘ Set the voltage. Connect the Agilent A voltage input to the output.
Agilent Technologies N5700 Series User Manual
Ensure that the load wiring insulation rating is greater than or equal to the maximum output voltage of the power supply. When a low voltage limit has been set, the instrument will ignore any programming commands that attempt to set the output voltage agulent the low voltage limit. Calibration mode must be enabled for the power supply to accept any other calibration commands.
A password is required avilent calibration mode is being enabled and the existing password is not 0. Page Index OUP The display first scrolls through the four segments of the IP address, followed by the six segments of the Ethernet EA address.
Verification and Calibration Appendix B Measurement Techniques Electronic Load Many of the test procedures require the use of a variable load capable of dissipating the required power. Check AC input voltage. Pins 2 and 3, which are connected internally, are the signal common.
Page 85 Programming Examples 6 ‘ Set the voltage. Page Index verification equipment Page 23 Installation 2 Install the shield after you have finished connecting the load wires.
This command sets the condition of the Service Request Enable Register. Instructions are given for performing n570 procedures from a controller over the GPIB. Is the AC power cord defective?
Agilent Technologies System DC N User Manual | pages
Source Commands Language Reference 5 Source Commands Source commands program the voltage, current, triggered, and protection functions. Page 50 IP address has been n7500.
Install the Setup utility on your computer.